1200 Degree Horizontal Tube Furnace Application Notes:
(1)Furnace tube pressure can not be higher than 0.02MPa
(2)As the internal pressure of the cylinder is high, so the gas into the furnace tube, the cylinder must be installed on the valve, the proposed purchase of the company pressure valve, the company valve range of 0.01MPa-0.1MPa, Will be more accurate and safe
(3)When the furnace temperature is higher than 1000 ℃, the furnace tube can not be in a vacuum state, the furnace pressure and atmospheric pressure equivalent to maintain the atmospheric pressure
(4)The gas flow into the furnace tube should be less than 200 SCCM to avoid the impact of cold airflow on the heated quartz tube.
Warranty: One year limited warranty with lifetime support, excluding heating element,cooling fans, thermocouple and ceramic tube damaged by user.
Attention: Any damages caused by the use of corrosive and acidic gases, and any damage from wrong operate and use are not under the coverage of One Year Limited Warranty.